TUF-2000H Ultrasonic Flowmeter was designed to measure rate of fluid flows in a closed line. Ultrasonic flowmeter does not contact to liquid and are attached on line with clamp. They are not get dirty and and maintenance is not required.Installation of these are easy and quick. TUF-2000H Ultrasonic Flowmeter works according to the transit time principle and are used two transducers as receive and transmitter.
Receive and transmitter transducers are mounted to outside of line depending on the distance between determined.Depending on diameter of pipe and feature of liquid transducers are placed according to any one method of W method, Vmethod or Z method. Between two transducers of Ultrasonic Flowmeter are available a specific sound frequency transmission.Transmitter transducer sends to sound frequency and receive transducer take to this sound frequency. This frequency signal transmission is provided by fluid in line between receive and transmitter.Transmitter sends signal, receive takes to this signal and as a result, a time difference occurs.This time difference is directly related to speed of fluid. This relationship is calculated by the following formula.
Θ : Angle in flow direction
M : Ultrasonic signal transmission coefficient
D : Pipe diameter
Tup : Transmission time to signal of first transducer
Tdawn : Transmission time to signal of second transducer